Magnetic float level gauge
The main components of the UHM-HIII magnetic float level gauge : sensor, magnetic float, transmitter.
Working principle: When the liquid level changes, the magnetic floating ball will also extend the pipe up and down, and the magnetic steel in the floating ball will first pick up the reed switch at the corresponding position in the sensor, forcing the total resistance (or voltage) of the sensor to occur. Change, and then convert the changed resistance (or voltage) signal into a 4-20 mA standard current signal transmission control room through the transmitter, and realize the remote monitoring and control of the liquid level in the container through the display instrument or DCS. .
Measuring range: 0 ~ 6000mm, greater than 6000mm requires special customization
Working pressure: -1.0～6.4Mpa
Supply voltage: 24V DC
Output signal: 4-20mA two-wire system
Accuracy: 0.5% F·S, 1.0% F·S
Ambient temperature: -70～60°C
Medium temperature: ≤150°C
Link flange: DN80 ~ DN200
Applicable medium: non-corrosive medium for PTFE
Explosion-proof mark: ExiaIICT6 Ga